<strong>Key Features:</strong>
<ul>
 	<li>Easy-to-use operating menu and color touch screen with selectable user modes, device statistics, icons, and function keys.</li>
 	<li>Comprehensive 22 parameter results including 3-part WBC differentiation in about 2 minutes.</li>
 	<li>Unique sample rotor to reduce risk of biohazard exposure.</li>
 	<li>Comprehensive QC package providing Levy-Jennings plots for 6 QC levels and a range of self-diagnostic functions for reliability and accuracy.</li>
 	<li>Clog detection and prevention with automatic system cleaning processes.</li>
</ul>
&nbsp;

<strong>Technical specifications:</strong>
<table width="651">
<tbody>
<tr>
<td width="191">Throughput</td>
<td width="459">30 tests/hour</td>
</tr>
<tr>
<td width="191">Parameters</td>
<td width="459">22 hematology parameters, including 3-part  differential: WBC, LYM, MON, GRA, LYM%, MON%,  GRA%, RBC, HCT, MCV, HGB, MCH, MCHC, PLT, PCT,  MPV, P-LCC, P-LCR, RDW-CV, RDW-SD, PDW-CV,  PDW-SD</td>
</tr>
<tr>
<td width="191">Measuring principle</td>
<td width="459">volumetric impedance method and light absorbance for HGB measurement</td>
</tr>
<tr>
<td width="191">Sample volume</td>
<td width="459">25 μl of whole blood</td>
</tr>
<tr>
<td width="191">Sampling method</td>
<td width="459">open tube system with automatic  sample rotor</td>
</tr>
<tr>
<td width="191">Calibration</td>
<td width="459">automatic or manual</td>
</tr>
<tr>
<td width="191">Quality control</td>
<td width="459">6 QC level range, SD and CV, Levey-Jennings  charts</td>
</tr>
<tr>
<td width="191">Data storage capacity</td>
<td width="459">10 000 results including RBC, PLT, WBC 3-part histograms and full  patient data</td>
</tr>
<tr>
<td width="191">Data input and output</td>
<td width="459">·       color touch screen

·       built-in thermal printer</td>
</tr>
<tr>
<td width="191">Dimensions &amp; Weight</td>
<td width="459">·       Dimensions: 32 × 37 × 27 сm

·       Weight: 10 kg</td>
</tr>
</tbody>
</table>
&nbsp;