<strong>Key Features:</strong>
<ul>
 	<li>Automatically pipetting and diluting of samples and reagents, incubation, shaking, plate washing, reading and output of results.</li>
 	<li>Dual (2) microplate design offers streamlined processing, reading and reporting of results.</li>
 	<li>Open architecture, fully customizable reporting and test designer allows assay configuration (down to minute detail).</li>
 	<li>Іntelligent sample rack, built-in patient-barcode scanner, and on-board camera for remote troubleshooting.</li>
 	<li>192 patient positions on convenient sliding sample-tray.</li>
 	<li>Simultaneous processing of 1 to 8 analytes in a single batch.</li>
 	<li>Compact footprint with high performance.</li>
</ul>
&nbsp;
<table>
<tbody>
<tr>
<td style="text-align: center;" width="209">Light source</td>
<td width="387">LED lamp</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Number of microplates</td>
<td width="387">2 plates (96 wells) and 1pre-dilution plate</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Number of samples</td>
<td width="387">192</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Number of reagents</td>
<td width="387">16</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Dispense system</td>
<td width="387">Single probe, dual needle</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Dispense volume</td>
<td width="387">1 up to 300 μl</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Photometer wavelengths</td>
<td width="387">405, 450, 550, 570 and 620 nm standard (other lengths available)</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Photometric range</td>
<td width="387">0 to 3,0 OD</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Precision</td>
<td width="387">0 to 2.000 OD: +‐0.003 OD or CV &lt; 1%, 2.001 OD to 3.000 OD: CV &lt; 1.5%</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Resolution</td>
<td width="387">0,001 OD</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Thermal control of plates</td>
<td width="387">up to 40 °C</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Shaker type</td>
<td width="387">orbital shaker with no-spill design</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Shaker parameters</td>
<td width="387">amplitude: 0,6 mm; adjustable speed: up to 900 RPMs; timer:  0 – 2h</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Data representation</td>
<td width="387">point-to-point, linear regression, cubic spline, 4PL, 5PL, Lin-Lin, Lin-Log, Log-Log</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Data output</td>
<td width="387">Lab System Integration (LSI) with Bi-directional</td>
</tr>
<tr>
<td style="text-align: center;" width="209">Dimensions &amp; Weight</td>
<td width="387">·       Dimensions: 64 × 45 × 57 сm

·       Weight: 28 kg</td>
</tr>
</tbody>
</table>
&nbsp;