<strong>Key Features:</strong>

<strong> </strong>
<ul>
 	<li>High throughput up to 200 tests/hour.</li>
 	<li>Direct chemiluminescence without enzymes for better stability.</li>
 	<li>Broad CLIA test menu with more than 160 assays.</li>
 	<li>Calibrators and internal controls are included in the reagent kits.</li>
 	<li>Single reaction cup instead of reaction module for avoiding increase cuvette utilization.</li>
 	<li>Continuous loading of reagents/samples without waiting or interrupting tests.</li>
 	<li>Accurate pipetting technologies include liquid level, clot, and crash detection</li>
 	<li>Intuitive real-time indicator light of reagent, sample, and consumables.</li>
 	<li>Bidirectional LIS connection.</li>
</ul>
&nbsp;

<strong>Technical specifications:</strong>

&nbsp;
<table>
<tbody>
<tr>
<td width="189">Chemiluminescence Principle</td>
<td width="453">Direct chemiluminescence, magnetic microbeads separation and key technology of ABEI-label</td>
</tr>
<tr>
<td width="189">Throughput</td>
<td width="453">Throughput: up to 200 tests/hour; 24 hours ready to use</td>
</tr>
<tr>
<td width="189">Reagent Loading</td>
<td width="453">20 reagent positions on board with continuous loading and 24-hour refrigerated</td>
</tr>
<tr>
<td width="189">Sample Loading</td>
<td width="453">Up to 72 sample positions with automatic barcode reading</td>
</tr>
<tr>
<td width="189">Calibration</td>
<td width="453">10 points master curve, 2-points re-calibration</td>
</tr>
<tr>
<td width="189">Reaction System</td>
<td width="453">Incubating 80 cuvettes simultaneously with five-sided heating at 37.0°C (±0.3°C)</td>
</tr>
<tr>
<td width="189">Sampling System</td>
<td width="453">Titanium needle with teflon coated for carry-over prevention; Liquid level detection, clot detection, prevent bubble interference, collision detection, wash automatically</td>
</tr>
<tr>
<td width="189">Washer</td>
<td width="453">Resuspension washing mode efficiently reduces non-specific binding and improves sensitivity</td>
</tr>
<tr>
<td width="189">Testing control</td>
<td width="453">Real-time status display for each test procedure</td>
</tr>
<tr>
<td width="189">LIS connection</td>
<td width="453">Bidirectional LIS connection</td>
</tr>
<tr>
<td width="189">Dimensions &amp; Weight</td>
<td width="453">·       Dimensions: 78 × 90 × 75 сm

·       Weight: 127 kg</td>
</tr>
</tbody>
</table>