<strong>Features:</strong>
<ul>
 	<li>Powerful reaction cuvette rinsing system for excellent accuracy and reproducibility.</li>
 	<li>Can be adapted to reagents from any manufacturer to save costs.</li>
 	<li>Features liquid level detection and needle collision protection.</li>
 	<li>Easy to use with built-in quality control software that allows you to display and print diagrams.</li>
</ul>
<strong>Technical Characteristics:</strong>
<table width="614">
<tbody>
<tr>
<td style="text-align: center;" width="245">Productivity</td>
<td width="369">300 tests/hour</td>
</tr>
<tr>
<td style="text-align: center;" width="245">System type</td>
<td width="369">open system</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Measurement parameters</td>
<td width="369">photometry, turbidimetry</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Methods</td>
<td width="369">End-point, Kinetic, two points, multi-standard, bichromatic, serum Blank, immunoturbidimetric etc.</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Source of the light</td>
<td width="369">halogen lamp</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Wavelength range</td>
<td width="369">340 – 810 nm</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Absorbency</td>
<td width="369">0.0~4.0A</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Absorbency characteristics</td>
<td width="369">precision: 0.0001A

Repetition of absorbency: coefficient variance (CV)≤1.0%</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Light filters</td>
<td width="369">10 pcs: 340, 405, 450, 505, 546, 578, 620, 670, 750, 800 нм</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Cuvettes</td>
<td width="369">90 reusable cuvettes (optical path - 6 mm)</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Reaction volume</td>
<td width="369">180-500 μl</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Reaction temperature</td>
<td width="369">37⁰C</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Cuvettes washing system:</td>
<td width="369">8-step reaction cuvette rinsing system</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Calibration</td>
<td width="369">Linear, nonlinear, one- point and multi-point.</td>
</tr>
<tr>
<td style="text-align: center;" width="245">&nbsp;

Quality control</td>
<td width="369">·       conducting a multi-level control test for each indicator

·       calculation of the coefficient of variation (CV in %)

·       building, displaying and printing a quality control diagram</td>
</tr>
<tr>
<td width="245">Operating system</td>
<td width="369">Windows-based software</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Printer</td>
<td width="369">connection of an external printer

(supplied with the analyser)</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Reagent volume</td>
<td width="369">·       Reagent 1 - 1-400 µl, 1 µl step

·       Reagent 2 - 1-400 µl, 1 µl step</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Reagents</td>
<td width="369">·       possibility of placing 40 reagents

·       constant cooling on board</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Sample positions</td>
<td width="369">60 items (including calibrator/standard, control, and STAT studies)</td>
</tr>
<tr>
<td style="text-align: center;" width="245">STAT positions (urgent)</td>
<td width="369">patient or control sample at any time</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Sample type</td>
<td width="369">serum, plasma, urine, cerebrospinal fluid, etc.</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Sample volume</td>
<td width="369">·       1-100 µl, 0,1 µl step

·       automatic sample dilution</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Sample/reagent dispensing and mixing</td>
<td width="369">·       2 hands for dispensing

·       1 independent mixer</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Dimensions</td>
<td width="369">85 х 72 х 53 sm</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Weight</td>
<td width="369">90 kg</td>
</tr>
</tbody>
</table>
&nbsp;