<strong>Key features</strong><strong>:</strong>
<ul>
 	<li>High accuracy of reactions and optical system. Spot photometry with high-speed digital data transmission system.</li>
 	<li>Optimal test sequence program to prevent instrument overload.</li>
 	<li>Flexibility of configuration for sample/reagent positions.</li>
 	<li>Displaying in real-time the working status of the sample, reagent drum and the reaction curve.</li>
</ul>
&nbsp;

<strong>Technical Characteristics:</strong>
<table width="614">
<tbody>
<tr>
<td style="text-align: center;" width="245">Productivity</td>
<td width="369">360 tests/hour without ISE (540 tests/hour with ISE)</td>
</tr>
<tr>
<td style="text-align: center;" width="245">ISE module (optional)</td>
<td width="369">K+, Na+, Cl-</td>
</tr>
<tr>
<td style="text-align: center;" width="245">&nbsp;

Measurement parameters</td>
<td width="369">
<ul>
 	<li>colourimetry</li>
 	<li>turbidimetry</li>
 	<li>ion-selective electrodes</li>
</ul>
</td>
</tr>
<tr>
<td style="text-align: center;" width="245">&nbsp;

Methods</td>
<td width="369">end-point, fixed time, kinetics, ISE (optional); single/double reagent chemistry, supporting single/double-wavelength mode</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Number of samples</td>
<td width="369">50 (can be increased due to the middle circle)</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Sample volume</td>
<td width="369">2-35 μl, step - 0.1 μl</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Number of reagents</td>
<td width="369">50 positions with cooling at a temperature of 2-12°C for 24 hours</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Reagent volume</td>
<td width="369">R1:100-200 μl, step - 0.5 μl

R2:10-200 μl, step - 0.5 μl</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Reaction volume</td>
<td width="369">100 - 360 μl</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Dosage of reagents</td>
<td width="369">one dosing arm for adding samples and reagents with liquid level determination function, and clot prevention function</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Reaction temperature</td>
<td width="369">37±0,3°C</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Light filters</td>
<td width="369">12 pcs: 340, 380, 412, 450, 505, 546, 570, 605, 660, 700, 740, 800 nm</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Source of the light</td>
<td width="369">halogen-tungsten lamp, 12V / 20W (service life - 2000 h)</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Cuvettes</td>
<td width="369">80 disposable cuvettes (optical path - 5 mm)</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Water consumption</td>
<td width="369">≤6,5 l / h</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Linear range</td>
<td width="369">0 – 3,5 units of OD</td>
</tr>
<tr>
<td style="text-align: center;" width="245">&nbsp;

Calibration</td>
<td width="369">K Factor, two-point linear, multi-point linear, Logit-Log 4P, Logit-Log 5P, exponential 5P, polynomial 5P, parabola, spline, Log 3P, broken line</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Dimensions</td>
<td width="369">86 х 66 х 55 cm</td>
</tr>
<tr>
<td style="text-align: center;" width="245">Weight</td>
<td width="369">115 ±15 kg</td>
</tr>
</tbody>
</table>
<p style="text-align: left;"></p>