<strong>Key Features:</strong>

&nbsp;
<ul>
 	<li>High throughput: up to 200 endpoint reactions per hour.</li>
 	<li>Automatically pipetting and diluting of samples and reagents, incubation, shaking, plate washing, reading and output of results.</li>
 	<li>The open programmable system provides great flexibility with the ability to customize parameters of analysis.</li>
 	<li>An unlimited number of analysis protocols.</li>
 	<li>Ready-made analysis protocols for Monobind immunoassy.</li>
 	<li>Simultaneous processing of 1 to 12 analytes with automated work-schedule.</li>
 	<li>Retention of built-in calibration curves and no need for replacement tips to reduce costs.</li>
</ul>
&nbsp;

<strong>Technical specifications:</strong>
<table width="651">
<tbody>
<tr>
<td style="text-align: center;" width="191">Light source</td>
<td width="459">tungsten-xenon lamp for ELISA and luminescent reader for CLIA</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Reading</td>
<td width="459">4-channel (simultaneous reading on four optical channels)</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Photometer wavelengths</td>
<td width="459">405, 450, 492, 630 nm</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Number of samples</td>
<td width="459">96 (including calibrators and controls)</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Number of reagents</td>
<td width="459">27 or 44 (provided they are placed in the sample area)</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Washing system</td>
<td width="459">automatic 8-probe</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Dispense volume</td>
<td width="459">2 μl – 1,95 ml</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Photometric range</td>
<td width="459">-0,2 to 3,0 A (ELISA); 106 RLU (CLIA)</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Precision</td>
<td width="459">± (1% of the reading +0.005A from 0 to 1.5A); ± (2% of the reading +0.005A from 1.5 to 3.0A)</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Thermal control of plates</td>
<td width="459">25°C, 37°C or room temperature</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Data representation</td>
<td width="459">single standard, factor, multi-calibrator point-to-point, linear regressions, log-logit, cubic spline, and nonlinear regressions (curve fit)</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Data output</td>
<td width="459">Lab System Integration (LSI)</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Dimensions &amp; Weight</td>
<td width="459">·       Dimensions: 86 × 51 × 40 сm

·       Weight: ≈ 35 kg</td>
</tr>
</tbody>
</table>
&nbsp;