<strong>Key Features:</strong>
<ul>
 	<li>29 parameters in less then 60 seconds.</li>
 	<li>Multi-patented led based flow cytometry and dynamic clustering for WBC differentiation.</li>
 	<li>Small sample volume and pre-dilution mode with automatic dosage of diluent.</li>
 	<li>Testing of samples up to 24 hours after collection without any affect.</li>
 	<li>Continuous loading and automatic rerun option.</li>
 	<li>Cost effectiveness due to low reagent consumption.</li>
 	<li>Flush Cleaner on board for automatic flushing.</li>
 	<li>User-friendly interface and large high-resolution color touch screen.</li>
 	<li>New counting chambers design allows faster service and lower maintenance cost.</li>
 	<li>Accurate counting of both oxyhemoglobin and carboxyhemoglobin with low influence of physiological methemoglobin on the measurement for correct patient diagnosis.</li>
</ul>
&nbsp;

<strong>Technical specifications:</strong>
<table width="651">
<tbody>
<tr>
<td width="191">Throughput</td>
<td width="459">70 tests/hour</td>
</tr>
<tr>
<td width="191">Parameters</td>
<td width="459">29 parameters:

·       WBC, RBC, PL Т, HGB, НТС, MCV, RDW-CV, RDW-SD, мен, мене, MPV, L УМ#, L УМ%, MON%, MON#, NEU%, NEU#, EOS%, EOSI*, BAS%, BAS#;

·       RUO: ALУ%, ALУ#, ІММ%, ІММ#, РСТ, PDW, P-LCR, Р–LCC.</td>
</tr>
<tr>
<td width="191">Sample volume</td>
<td width="459">whole blood 17,5 µl +pre-dilution mode</td>
</tr>
<tr>
<td width="191">Sampling method</td>
<td width="459">open tubes and autoloader for closed tubes</td>
</tr>
<tr>
<td width="191">Autoloader</td>
<td width="459">·       10 racks - 5 samples (50 samples in total);

·       continuous loading;

·       automatic rerun option;

·        different type of tubes.</td>
</tr>
<tr>
<td width="191">Linearity</td>
<td width="459">·       WBC  0,2 - 100 (10<sup>3</sup>/µl)

·       RBC 0,02 - 8 (10<sup>6</sup>/µl)

·       HGB  0,2 - 24 (g/µl)

·       PLТ 10 - 2000 (10<sup>3</sup>/µl)</td>
</tr>
<tr>
<td width="191">Precision (CV)</td>
<td width="459">·       WBC &lt;25%

·       RBC &lt;2%

·       HGB &lt;1.5%

·        PLТ &lt;5%</td>
</tr>
<tr>
<td width="191">Data storage capacity</td>
<td width="459">·       35 000 results

·       500 system events or errors

·        60 reagents entries</td>
</tr>
<tr>
<td width="191">Connection</td>
<td width="459">·       7 USB ports;

·       internet connection;

·       RS232;

·       bidirectional LIS-connection.</td>
</tr>
<tr>
<td width="191">Accessories</td>
<td width="459">·       barcode reader;

·       keyboar;

·       printer.</td>
</tr>
<tr>
<td width="191">Dimensions &amp; Weight</td>
<td width="459">·       Dimensions: 43 × 50,7 × 54 сm

·       Weight: 31 kg</td>
</tr>
</tbody>
</table>
&nbsp;