<strong>Key Features:</strong>
<ul>
 	<li>Сompact design for easy use in any type of laboratory.</li>
 	<li>An excellent optics and performance provide reading a 96-well plate in about 8 seconds.</li>
 	<li>The open programmable system provides great flexibility with the ability to customize parameters of analysis.</li>
 	<li>An unlimited number of analysis protocols. Ready-made analysis protocols for Monobind immunoassy.</li>
 	<li>Curve plotting and editing. Stored curve and two point calibration options for reagent savings.</li>
 	<li>Program and read any number of assays per plate.</li>
</ul>
&nbsp;

<strong>Technical specifications:</strong>
<table width="651">
<tbody>
<tr>
<td style="text-align: center;" width="191">Light source</td>
<td width="459">remote halogen lamp with saver features</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Reading</td>
<td width="459">8-channel reader</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Photometer wavelengths</td>
<td width="459">405, 450, 492, 630 nm plus optional</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Measurement Range</td>
<td width="459">-0.20 to 4.5 or above optical density</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Photometric Accuracy</td>
<td width="459">+/- (1% +/- 0.010) 0.0 до 1.5 ABS

+/- (2% +/- 0,010) 1.5 до 3.0 ABS</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Stability</td>
<td width="459">drift of no more than 0.005A in 8 hours</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Microplate type</td>
<td width="459">96-well microplate or strips</td>
</tr>
<tr>
<td style="text-align: center;" width="191">&nbsp;

Data representation</td>
<td width="459">point-to-point, linear regressions, and dose response modes and

protocol types such as immunometric, competitive, cutoff and

raw data.</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Data output</td>
<td width="459">Lab System Integration</td>
</tr>
<tr>
<td style="text-align: center;" width="191">Dimensions &amp; Weight</td>
<td width="459">·       Dimensions: 18,5 × 18,5 × 42 сm

·       Weight: 4,5 kg</td>
</tr>
</tbody>
</table>
&nbsp;