Productivity
120 tests/hour without ISE (400 tests/hour with ISE)
ISE module (optional)
K +, Na +, Cl-
Measurement parameters
absorbing photometry, turbidimetry, ion-selective electrodes
Methods
end-point, fixed time, kinetics, ISE (optional); single/double reagent chemistry; monochromatics/biochromatics; linear/nonlinear multipoint calibration
Number of samples
40 (can be increased to 80 due to the middle circle)
Sample volume
2-45 μl, step - 0.1 μl
Number of reagents
40 positions with cooling at a temperature of 2-12°C for 24 hours
Reagent volume
10-250 μl, step - 0.5 μl
Reaction volume
100 - 360 μl
Dosage of reagents
one dosing arm for adding samples and reagents with liquid level determination function, and clot prevention function
Reaction temperature
37 ± 0.3 ° C
Light filters
8 pcs: 340, 405, 450, 510, 546, 578, 630, 670 nm
Source of the light
halogen-tungsten lamp, 12V / 20W (service life - 2000 h)
Cuvettes 
40 disposable cuvettes (optical path - 5 mm)
Water consumption
≤3 l / h
Linear range
0 - 4.0 units of OD
Calibration
linear (single-point, two-point and multi-point), Logit-Log 4P, Logit-Log 5P, spline, exponential, polynomial, parabola
Dimensions, weight
69.5 x 58.5 x 60 cm; ≤130 kg