Productivity
360 tests / hour (540with ISE-module)
Methods of analysis
endpoint, fixed time, kinetics, moni/bireagent chemistry, monochromatic/bichromatic, linear/nonlinear multipoint calibration.
ISE module
Optional, parameters K + / Na + / Cl-
The number of sample positions
50 (can be increased by the middle circle)
The number of reagent positions
50 (can be increased by the middle circle) with cooling at a temperature of 2-12 ° C for 24 hours
Sample volume
2-35 μl, step 0.1 μl
Reagent volume
R1: 100-200 μl, step - 0.5 μl; R2: 10-200 μl, step 0.5 μl
Dosing of reagents
One dosing arm for adding a sample and reagent with the function of determining the liquid level, and the function of preventing the clots formation
The reaction volume
100-360 μl
Wavelength
12 pcs: 340, 380, 412, 450, 505, 546, 570, 605, 660, 700, 740, 800 nm
Cuvettes
80 reusable cuvettes
Washing station
8-step washing of cuvettes; water consumption ≤ 6.5 l / h
Dimensions, weight
860 x 660 x 550 mm; 115 ± 15 kg