Light source
halogen, lamp saver feature
Wavelengths
405, 450, 492, 630nm and additional filter upon request
Photometric Accuracy
+/- (1% +/- 0.010) 0.0 through 1.5 ABS, +/- (2% +/- 0,010) 1.5 through 3.0 ABS
Optical Measurement Range
-0.20 to 4.0 Absorbance units and above
Stability
drift of no more than 0.005A in 8 hours
Optical system
8-channel 
Vessel
standard 96 well microtiter plates or strip trays
Data output
LIS-connect possibility
QC options
store control data, print Levey-Jennings or QC range plots, calculate SDs
Power supply
100 - 240VAC, 50-60Hz
Environmental Conditions
tº: 5…40°C; Humidity : 40…80%
Dimensions
19 x 20.3 x 42cm
Weight
5,7 kg