Productivity
260 tests / h (420 with ISE)
Methods of analysis
End point, kinetics, fixed time, 1-2 reagent, multistandard, bichromatic, blank reagent/serum, immunoturbidimetry
ISE
Optional
Number of sample positions
60 sample positions (including the standard, blank and control, serum and output tube are available)
Number of reagent positions
40 reagent positions (80 reagent positions are optional)
Sample volume
1-100 μl, step 0.1 μl
The volume of the reagent
R1: 1-400 μl, step 0.1 μl  R2: 1-400 μl, step 0.1 μl
Wavelength range
340 – 810 nm
Installed light filters
10 basic 340 nm, 405 nm, 450 nm, 505 nm, 546 nm, 578 nm, 620 nm, 670 nm, 750 nm, 800 nm and optional: 380 nm, 492 nm, 630 nm, 650 nm, 660 nm, 700 nm,810 nm 810 nm
Dimensions, weight
850 mm x 720 mm x 530 mm; 90.0 kg