Number of measuring channels
Four channels. 
Productivity
up to 120 PT tests/hour
The principle of measurement
Metering system is photometric, with automatic start function
The volume of the reaction mixture
The range of working volumes: min 5 μl - max. 200 μl
Data storage
Storing more than 1,000,000 measurement data and corresponding curves according to identification number, date and time of measurement
Preparation of samples
Built-in incubator 37.4°C +/- 0.4°, 60 positions for samples, 14 positions for reagents, 5 STAT positions 
Sampling method
Sampling system from open and closed trials (special model of analyzer)
Calibration
6-point calibration curve, automatic or manual calibration
Data entry
Liquid crystal touchscreen display, USB-barcode scanner
P printer
USB interface for connecting an external printer
Quality control
Possibility of quality control (QC)
Dimensions, weight
850x750x510mm; 80 kg